My green tile values are not correct in the EasyMatch QC software. How can I change them?
To edit the existing green tile values for your instrument in EasyMatch QC go to Sensor/ Diagnostics/ Green Tile Test. An EasyMatch QC window will open that prompts you to install the standard port plate. (The standard port plate is the one with the largest area view possible for your instrument.)