The current, available ASTM D1003 Haze Standards have nominal Haze% values of 1, 5, 10, 20 and 30 with air (transparent solids) or the transmission cell filled with DI water being 0 (transparent liquids). Here are some thoughts on further options. Continue reading
HunterLab sphere instruments with CIE d/8 geometries conform to the requirements of ASTM D1003 Section 8: Procedure B Spectrophotometer. The measurement of transmission haze using Procedure B instruments will be in close agreement with ASTM D1003 Procedure A Haze meter.
Here is a more detailed description of how the HunterLab UltraScan PRO, UltraScan VIS and ColorQuest XE meet the requirements of Section 8.
FAQ: “We have a scientist here who is looking at films containing small bubbles and attempting to do measure them by measuring Haze% on a HunterLab spectrophotometer. I suspect her problem in obtaining numbers that “make sense” from what you get from just eyeballing the samples (and trying to look through them at the outlines of a object behind the film to gage the amount of regular transmittance) is that her samples have Haze% > 30 %, which means that a spectrophotometer (or even a haze meter) isn’t the appropriate tool to use for the analysis.
How is transmission haze measured at levels > 30%?”