ERIC stands for Effective Residual Ink Concentration and is a control parameter used to quantify the amount of residual ink left in recycled or de-inked pulp (DIP) and paper before and after de-inking and/or bleaching processes. Other parameters measured are brightness and dirt content.
Absorption curve showing residual ink levels at NIR 800 to 1200 nm.
Based on joint research by PAPRICAN (Pulp And Paper Research Institute of Canada) and Technidyne Corporation, it was determined that there was a relationship between the ink and only the ink (not dyes or colorants, paper lignins, pulp yellowing or loss of brightness due to bleaching or process effects) and the reflectance in the 800 – 1200-nm infrared region of the spectrum. This unique correlation makes the ERIC value based on absorption at 950 nm ideal for verification of residual ink concentration for recycled paper.
The ERIC value is affected by the distribution of ink particle size with this metric being most effective for residual ink determination when the ink particles are submicron in size.
ERIC values in ppm = (Ink Absorption Coefficient * Absorption at 950-nm)
where the Ink Absorption Coefficient is based on analytical determination.
ERIC value is a dimensionless ratio of the light absorption coefficient of pulp or paper containing ink to the light absorption coefficient of the ink itself, both being determined at a wavelength of 950 nm. The User will need to calibrate the ERIC value to report in ppom with an added Ink Absorption Coefficient determined by the User.
Jordan, B.D. and Popson S.J., 2nd Research Forum on Recycling, St. Adele, Proceedings: 153-169 (1993)
Jordan, B. D. and Popson, S.J. “Measuring the concentration of residual ink in recycled newsprint”, Journal of Paper Science 20 (6):J161-167 (1994)
Trepanier, R.J., Jordan, B., Nguyen, N. and Patschka, H.J. “High-Magnification Image Analysis with Novel Background Reflectance Technique for Measuring Residual Ink in Sheets,” Journal of Pulp and Paper Science, 23 (3): J129 (1997)
Jordan, B. and O’Neill, M. “The Kubelka-Munk Absorption Coefficients of Several Carbon Blacks and Water-Based Printing Inks,” Journal of Pulp and Paper Science, 20 (12): 371 (December 1994)
Carre, B., Galland, G. and Saint Amand, F.J. “Control of Detachment and Removal of Ink by Image Analysis,” Centre Technique de l’Industrie des Papiers (CTP), Grenoble, France, doc. # 1670, (9 March 1994)
TAPPI Test Method T 567 Determination of Effective Residual Ink Concentration by Infrared Reflectance Measurement
ISO 22754 Pulp and paper — Determination of the effective residual ink concentration (ERIC number) by infrared reflectance measurement
PAPTAC E.8 ERIC value for residual ink concentration
Contrast Ratio Opacity measurement quantifies how opaque a sheet sample using two readings. One reading is taken with the black glass or light trap backing the sample at the port. The second reading is taken with the sample backed by a white tile or the white sample clamp insert. EasyMatch QC will prompt you for the proper backing when taking opacity measurements.
FAQ: “… I just need to find a way to better condition the samples for consistency. I can get some pretty consistent results on my own, but once it’s in a general QC lab, it’s going to need some standardization.”